Human LTA ELISA Kit For the quantitative in vitro determination of Human Lung tumor Antigen concentrations in serum - plasma - celiac fluid - tissue homogenate - body fluid FOR LABORATORY RESEARCH USE ONLY. NOT FOR USE IN DIAGNOSTIC PROCEDURES. This package insert must be read in its The ensemble before using this product. The ELISA ENZYME LINKED IMMUNOSORBENT ASSAYINTENDED USE AND TEST PRINCIPLEThis LTA ELISA kit is intended Laboratory for Research use only and is not for use in diagnostic or therapeutic procedures. The Stop Solution changes the color from blue to yellow and the intensity of the In order to measure the concentration of LTA in the sample, this LTA ELISA Kit includes a set of calibration standards. The calibration standards are assayed at the same time as the samples and allow the operator to Produce a standard curve of Optical Density versus LTA concentration. The concentration of LTA in the samples is then determined by com Paring the OD of the samples to the standard curve. SAMPLE COLLECTION AND STORAGESSerum - Use a serum separator tube and allow samples to clot for 2 hours at room temperature or overnight at 4Â°C before centrifugation for 20 minutes at approximately 2000Ã—g. And assay immediately or aliquot and store samples at -20Â°C. Avoiding freeze-thaw cyclesPlasma - Collect plasma using heparin as an anticoagulant. Centrifuge samples for 30 minutes at 2000Ã—g at 2-8Â°C within 30 minutes of collection. At -20Â°C. Avoiding freeze-thaw cycles.Cell culture supernates, tissue homogenate and other biological fluids - Remove particulates by centrifugation and assay immediately or aliquot and store samples at -20Â°C. Samples shoule be centrifugated dequately and no hemolysis or granule was allowed.MATERIALS REQUIRED BUT NOT SUPPLIED1. 37 Â°C incubator2. Standard microplate reader capable of measuring absorbance at 450 nm3. P Recipe pipettes, disposable pipette tips and Absorbent paper4. Distilled or deionized waterREAGENTS PROVIDED All reagents provided are stored at 2-8Â°C. Refer to the expiration date on the label. Name 96 determinations 48 determinationsMicroelisa stripplate 12*8strips 12*4stripsStandard(6 Vial) 0.5ml/vial 0.5ml/vialSample diluent 6.0ml 3.0mlHRP-Conjugate reagent 10.0ml 5.0ml20X Wash solution 25ml 15mlChromogen Solution A 6.0ml 3.0mlChromogen Solution B 6.0ml 3.0mlStop Solution 6.0ml 3.0mlClosure plate membrane 2 2User manual 1 1Sealed Bags 1 1Note: 1. Standard concentration was followed by: 4000, 2000, 1000, 500, 250, 125 pg/ml.2. If samples generate values â€‹â€‹higher than the highest standard, please dilute the samples with Sample Diluent and repeat the assay .PRECAUTIONS1. Do not substitute reagents from one kit lot to another. Standard, conjugate and microtiter plates are matched for optimal performance. Use only the reagents supplied by manufacturer. 2. Allow kit reagents an d materials to reach room temperature (20-25Â°C) before use. Do not use water baths to thaw samples or reagents. 3. Do not use kit components beyond their expiration date. 4. Use only deionized or distilled water to dilute reagents 7. Do not remove microtiter plate from the storage bag until needed. Unused strips should be stored at 2-8Â°C in their pouch with the desiccant provided. 6. Use fresh disposable pipette tips for each transfer to avoid contamination. Disposable knuckle must be during the procedure procedure, since no known test method can offer complete assurance that products derived from Human Blood will not transmit infectious agents. Therefore, all blood derivatives should be considered potentially infectious and good laboratory practices should be followed. 9. All samples should be disposed in a manner that will ina 10. Waste Waste: be sodium hypochlorite to a final concentration of 1.0%. The waste should be allowed to stand for a minimum of 30 minutes to inactivate the viruses before disposal. 11. Substrate Solution is easily contaminated. If bluish prior Substrate B contain 20% acetone, keep this reagent away from sources of heat or flame. 13. Remove all kit reagents from refrigerator and allow them to reach room temperature (20-25Â°C). REAGENT PREPARATION AND STORAGEWash Solution (1X) - Dilute 1 volume of Wash solution (20X) with 19 volumes of deionized or distilled water. Wash Solution is stable for 1 month at 2-8Â°C. ASSAY PROCEDURE1. Prepare all reagents before starting assay Add to the micro-lisa stripplate.2. Add 50Î¼l of Standard or Sample to the appropriate wells Standard wells and sample wells e Xcept the blank well, cover with an adhesive strip and incubate for 60 minutes at 37Â°C.4. Wash the Microtiter Plate 4 times. Manual Washing - Remove incubation mixture by aspirating contents of the plate into a sink or proper waste container. a squirt bottle, fill each well completely with Wash Solution (1X), then aspirate contents of the plate into a sink or proper waste container. Repeat this procedure for a total of four times. After final wash, invert plate, and blot dry by Note: Hold the sides of the plate frame firm when washing the plate to assure that all strips remain securely in frame. Automated Washing - Aspirate all wells, then wash plates four times using Always adjust your brush to aspirate as much liquid as possible and set fill volume at 350Î¼L/well/wash. After final wash, invert plate, and blot dry by hitting plate mount absorbent paper or paper blanket until no moist Ure appears.5. Add chromogen solution A 50Î¼l and chromogen solution B 50Î¼l to each well. Gently mix and incubate for 15 minutes at 37Â°C. Protect from light.6. Add 50Î¼l Stop Solution to each well. The color in the wells Read the Optical Density (OD) at 450 nm using a microtiter plate reader The standard curve is generated by plotting the average OD (450 nm) obtained for each of the six standard concentrations on the vertical (X) 2. First, calculate the mean OD value for each standard and sample. All OD Values â€‹â€‹are subtracted by the mean value of the balnk well before result interpretation. Construct the standard curve using Graph paper To determine the amount in each sample, first locate the OD value on the Y-axis and extend a horizontal line to the standard curve. At the point of intersection, draw a vertical line to the X-axis and 4. Any variation in operator, pipetting and washing technique, incubation time or temperature, and kit age can cause variation in result. Each user should obtain their own standard curve.5. Intra-assay CV(%) and Inter-assay CV (%) are less than 15%.6. Assay range: 125 pg/ml â€“ 4000 pg/ml.7. Sensitivity: The minimum detectable dose of Human LTA is typically less than 100 pg/ml.8. No significant cross-reactivity or interference was observed. 9. Storage: 2-8 Â° C (Use frequently); six months (-20 Â° C). 10. QUOTE READ ONH;
Stone Bath Mat
Diatomaceous earth is very popular material these years.
Diatoms are tiny, single-celled algae found in plankton in ocean. Diatoms accumulated after they died, then fossilized into diatomaceous earth after hundreds of millions of years. The main components are silicon dioxide, light and porous. Electron microscope shows that diatomaceous earth is nanoscale porous material with diameter at 0.1-0.2 micrometers, the porosity is up to 90% and the pores are regularly organized into circles and its micropores quantity in unit area is several thousands times of active carbon's.
Such unique molecular characteristics contributed to the excellent function of diatomaceous earth.
Stone Bath Mat,Foot Bath Mat,Fast Absorbent Carpe,Absorbent Bath Mat
zhangjiagang wellyoung material co.,ltd , https://www.wellyoung-board.com